Freestyle Libre 2¶
The Freestyle Libre 2 system can automatically report dangerous blood glucose levels. The Libre2 Sensor sends the current blood sugar level to a receiver (reader or smartphone) every minute. The receiver triggers an alarm if necessary. With a self-modified LibreLink app and the xDrip+ app, you can continuously receive and display your blood sugar level on your smartphone.
The sensor can be calibrated in the range of -40 mg/dl to +20 mg/dl (-2,2 mmol/l to +1,1 mmol/l) to adjust differences between finger prick measurements and sensor readings.
BG readings can also be done using a BT transmitter like with the Libre1.
Step 2: Nainstalujte a nastavte aplikaci xDrip+¶
The blood sugar values are received on the smartphone by the xDrip+ App.
- If not already set up then download xDrip+ app and install one of the latest nightly builds from here.
- In xDrip+ select “Libre2 (patched App)” as data source
- If necessary, enter “BgReading:d,xdrip libre_receiver:v” under Less Common Settings->Extra Logging Settings->Extra tags for logging. This will log additional error messages for trouble shooting.
- In xDrip+ go to Settings > Interapp Compatibility > Broadcast Data Locally and select ON.
- In xDrip+ go to Settings > Interapp Compatibility > Accept Treatments and select OFF.
- to enable AAPS to receive blood sugar levels (version 2.5.x and later) from xDrip+ please set Settings > Interapp Settings > Identify Receiver “info.nightscout.androidaps”
- If you want to be able to use AndroidAPS to calibrate then in xDrip+ go to Settings > Interapp Compatibility > Accept Calibrations and select ON. Můžete také zkontrolovat v xDripu nastavení v částí Nastavení > Méně častá nastavení > Rozšířené kalibrace.
Step 3: Start sensor¶
In xDrip+ start the sensor with “Start Sensor” and “not today”.
In fact this will not physically start any Libre2 sensor or interact with them in any case. This is simply to indicate xDrip+ that a new sensor is delivering blood sugar levels. If available, enter two bloody measured values for the initial calibration. Now the blood glucose values should be displayed in xDrip+ every 5 minutes. Skipped values, e.g. because you were too far away from your phone, will not be backfilled.
After a sensor change xDrip+ will automatically detect the new sensor and will delete all calibration data. You may check you bloody BG after activation and make a new initial calibration.
Step 4: Configure AndroidAPS (for looping only)¶
- V AndroidAPS přejděte na kartu Konfigurace > Zdroj glykémie a vyberte ‘xDrip+’
- If AndroidAPS does not receive BG values when phone is in airplane mode, use Identify receiver as describe on xDrip+ settings page.
Ani v současné době není možné při používání Libre 2 jako zdroje glykémie povolit v rámci algoritmu SMB funkce ‘Vždy povolit SMB’ a ‘Povolit SMB po jídle’. Hodnoty glykémií z Libre 2 nejsou dostatečné vyhlazené, aby bylo použití těchto funkcí bezpečné. Další podrobnosti viz Vyhlazování glykémií.
Experiences and Troubleshooting¶
The connectivity is extraordinarily good. With the exception of Huawei mobile phones, all current smartphones seem to work well. The reconnect rate in case of connection loss is phenomenal. The connection can break off if the mobile phone is in the pocket opposite the sensor or if you are outdoors. When I am gardening, I use to wear my phone on the sensor side of my body. In rooms, where Bluetooth spreads over reflections, no problems should occur. If you have connectivity problems please test another phone. It may also help to set the sensor with the internal BT antenna pointing down. The slit on the applicator must be pointing down when setting the sensor.
Value smoothing & raw values¶
Technically, the current blood sugar value is transmitted to xDrip+ every minute. A weighted average filter calculates a smoothed value over the last 25 minutes. This is mandatory for looping. The curves look smooth and the loop results are great. The raw values on which the alarms are based jitter a little more, but correspond to the values that the reader also displays. In addition, the raw values can be displayed in the xDrip+ graph in order to be able to react in time to rapid changes. Please switch on Less Common Settings > Advanced Settings for Libre2 > “show Raw values” and “show Sensors Infos”. Then the raw values are additionally displayed as small white dots and additional sensor info is available in the system menu.
The raw values are very helpful when the blood sugar is moving fast. Even if the dots are jumpier you would detect the tendency much better as using the smoothed line to make proper therapy decisions.
The sensor runtime is fixed to 14 days. The 12 extra hours of Libre1 no longer exist. xDrip+ shows additional sensor information after enabling Advanced Settings for Libre2 > “show Sensors Infos” in the system menu like the starting time. The remaining sensor time can also be seen in the patched LibreLink app. Either in the main screen as remaining days display or as the sensor start time in the three-point menu->Help->Event log under “New sensor found”.
A sensor exchange takes place on-the-fly: Set new sensor shortly before activation. As soon as xDrip+ receives no more data from the old sensor, start the new sensor with the patched app. After one hour new values should appear automatically in xDrip+.
If not, please check the phone settings and proceed as with the first start. You have no time limit. Try to find the correct settings. No need to immediately replace the sensor before you tried different combinations. The sensors are robust and try permanently to establish a connection. Please take your time. In most cases you accidentally changed one setting which causes now problems.
Once successful please select “Sensor Stop” and “Delete calibration only” in xDrip. This indicates for xDrip+ that a new sensor is releasing blood sugar levels and the old calibrations are no longer valid and therefore have to be deleted. No real interaction is done with the Libre2 sensor here! You do not need to start the sensor in xDrip+.
You can calibrate the Libre2 with an offset of -40 mg/dl to +20 mg/dL [-2,2 mmol/l to +1,1 mmol/l] (intercept). The slope isn’t changeable as the Libre2 is much more accurate compared to the Libe1. Please check by fingerpricking early after setting a new sensor. It is known that there can arise big differences to the blood measurements. To be on the safe side, calibrate every 24 - 48 hours. The values are accurate up to the end of the sensor and do not jitter as with the Libre1. However, if the sensor is completely off, this will not change. The sensor should then be replaced immediately.
The Libre2 sensors contain plausibility checks to detect bad sensor values. As soon as the sensor moves on the arm or is lifted slightly, the values may start to fluctuate. The Libre2 sensor will then shut down for safety reasons. Unfortunately, when scanning with the App, additional checks are made. The app can deactivate the sensor even though the sensor is OK. Currently the internal test is too strict. I have completely stopped scanning and haven’t had a failure since then.
Time zone travelling¶
In other time zones there are two strategies for looping:
- leave the smartphone time unchanged and shift the basal profile (smartphone in flight mode) or
- delete the pump history and change the smartphone time to local time.
Method 1. is great as long as you don’t have to set a new Libre2 sensor on-site. If in doubt, choose method 2., especially if the trip takes longer. If you set a new sensor, the automatic time zone must be set, so method 1. would be disturbed. Please check before, if you are somewhere else, you can run otherwise fast into problems.
Altogether it is one of the smallest CGM systems on the market. Small, no transmitter necessary and mostly very accurate values without fluctuations. After approx. 12 hours running-in phase with deviations of up to 30 mg/dl (1,7 mmol/l)the deviations are typical smaller than 10 mg/dl (0,6 mmol/l). Best results at the rear orbital arm, other setting points with caution! No need to set a new sensor one day ahead for soaking. That would disturb the internal leveling mechanism.
There seem to be bad sensors from time to time, which are far away from the blood values. It stays that way. These should be immediately replaced.
If the sensor moved a little bit on the skin or is lifted somehow this can cause bad results. The filament which sits in the tissue is a little bit pulled out of the tissue and will measure different results then. Mostly probably you will see jumping values in xDrip+. Or the difference to the bloody values change. Please replace the sensor immediately! The results are inaccurate now.
Using bluetooth transmitter and OOP¶
Bluetooth transmitter can be used with the Libre2 with the latest xDrip+ nightlys and the Libre2 OOP app. You can receive blood sugar readings every 5 minutes as well as with the Libre1. Please refer to the miaomiao website to find a description. This will also work with the Bubble device and in the future with other transmitter devices. The blucon should work but has not been tested yet.
Old Libre1 transmitter devices cannot be used with the Libre2 OOP. They need to be replaced with a newer version or have a firmware upgrade for proper operation. MM1 with newest firmware is unfortunately not working yet - searching for root cause is currently ongoing.
The Libre2 OOP is creating the same BG readings as with the original reader or the LibreLink app via NFC scan. AAPS with Libre2 do a 25 minutes smoothing to avoid certain jumps. OOP generates readings every 5 minutes with the average of the last 5 minutes. Therefore the BG readings are not that smooth but match the original reader device and faster follow the “real” BG readings. If you try to loop with OOP please enable all smoothing settings in xDrip+.
The Droplet transmitter is working with Libre2 also but uses an internet service instead. Please refer to FB or a search engine to get further information. The MM2 with the tomato app also seems to use an internet service. For both devices you have to take care to have a proper internet connection to get your BG readings.
Even if the patched LibreLink app approach is smart there may be some reasons to use a bluetooth transmitter:
- the BG readings are identical to the reader results
- the Libre2 sensor can be used 14.5 days as with the Libre1 before
- 8 hours Backfilling is fully supported.
- get BG readings during the one hour startup time of a new sensor
Remark: The transmitter can be used in parallel to the LibreLink app. It doesn’t disturb the patched LibreLink app operation.
Remark #2: The OOP algorithm cannot be calibrated yet. This will be changed in the future.