아큐첵 인사이트 펌프(Accu-Chek Insight Pump)¶
이 소프트웨어는 DIY 인공 췌장 솔루션의 일부이며 판매용 제품이 아닙니다. 하지만 사용법을 읽고 배우고 이해 하셔야합니다. 이 소프트웨어는 당뇨관리를 대신 해주는 것은 아니지만, 시간을 들여 숙지하신 후 용하신다면 당뇨관리를 용이하게 도와주며 삶의 질을 향상시키는데 큰 도움을 줍니다. 성급하게 생각하지 마시고, 사용법을 숙지하시기를 권장합니다. 서용자가 소프트웨어 활용과 관련하여 대한 책임을 집니다.
주의: 만약 이 전에 Insight를 SightRemote와 같이 사용하셨을 경우, 최신 AAPS버전으로업데이트 해주시고 SightRemote를 삭제해주시기 바랍니다.¶
하드웨어와 소프트웨어의 필요 요건¶
- A Roche Accu-Chek Insight pump (any firmware, they all work)
Note: AAPS will write data always in first basal rate profile in the pump.
- An Android phone (Basically every Android version would work with Insight, but check on the Module page which Android version is required to run AndroidAPS.)
- The AndroidAPS app installed on your phone
The Insight pump should only be connected to one device at a time. If you have previously used the Insight remote control (meter), you must remove the meter from the paired devices list of your pump: Menu > Settings > Communication > Remove device
In Config builder of the AndroidAPS app select Accu-Chek Insight in the pump section
Tap the cog-wheel to open Insight settings.
In settings, tap on the button ‘Insight pairing’ at the top of the screen. You should see a list of all nearby bluetooth devices (below left).
On the Insight pump, go to Menu > Settings > Communication > Add Device. The pump will display the following screen (below right) showing the serial number of the pump.
Going back to your phone, tap on the pump serial number in the list of bluetooth devices. Then tap on Pair to confirm.
Both the pump and phone will then display a code. Check that the codes are the same on both devices and confirm on both the pump and the phone.
Success! Pat yourself on the back for successfully pairing your pump with AndroidAPS.
To check all is well, go back to Config builder in AndroidAPS and tap on the cog-wheel by the Insight Pump to get into Insight settings, then tap on Insight Pairing and you will see some information about the pump:
Note: There will be no permanent connection between pump and phone. A connection will only be established if necessary (i.e. setting temporary basal rate, giving bolus, reading pump history...). Otherwise battery of phone and pump would drain way too fast.
Note : It is now possible (only with AAPS v2.7.0 and above) to use ‘Always use basal absolute values’ if you want to use Autotune with Insight pump, even if ‘sync is enabled’ with Nightscout. (In AAPS go to Preferences > NSClient > Advanced Settings).
In the Insight settings in AndroidAPS you can enable the following options:
“Log reservoir changes”: This will automatically record an insulin cartridge change when you run the “fill cannula” program on the pump.
“Log tube changes”: This adds a note to the AndroidAPS database when you run the “tube filling” program on the pump.
“Log site change”: This adds a note to the AndroidAPS database when you run the “cannula filling” program on the pump. Note: A site change also resets Autosens.
“Log battery changes”: This records a battery change when you put a new battery in the pump.
“Log operating mode changes”: This inserts a note in the AndroidAPS database whenever you start, stop or pause the pump.
“Log alerts”: This records a note in the AndroidAPS database whenever the pump issues an alert (except reminders, bolus and TBR cancellation - those are not recorded).
“Enable TBR emulation”: The Insight pump can only issue temporary basal rates (TBRs) up to 250%. To get round this restriction, TBR emulation will instruct the pump to deliver an extended bolus for the extra insulin if you request a TBR of more than 250%.
Note: Just use one extended bolus at a time as multiple extended boluses at the same time might cause errors.
“Disable vibrations on manual bolus delivery”: This disables the Insight pump’s vibrations when delivering a manual bolus (or extended bolus). This setting is available only with the latest version of Insight firmware (3.x).
“Disable vibrations on automated bolus delivery”: This disables the Insight pump’s vibrations when delivering an automatic bolus (SMB or Temp basal with TBR emulation). This setting is available only with the latest version of Insight firmware (3.x).
“Recovery duration”: This defines how long AndroidAPS will wait before trying again after a failed connection attempt. You can choose from 0 to 20 seconds. If you experience connection problems, choose a longer wait time.
Example for min. recovery duration = 5 and max. recovery duration = 20
no connection -> wait 5 sec.retry -> no connection -> wait 6 sec.retry -> no connection -> wait 7 sec.retry -> no connection -> wait 8 sec....retry -> no connection -> wait 20 sec.retry -> no connection -> wait 20 sec....
“Disconnect delay”: This defines how long (in seconds) AndroidAPS will wait to disconnect from the pump after an operation is finished. Default value is 5 seconds.
For periods when pump was stopped AAPS will log a temp. basal rate with 0%.
In AndroidAPS, the Accu-Chek Insight tab shows the current status of the pump and has two buttons:
“Refresh”: Refreshes pump status
“Enable/Disable TBR over notification”: A standard Insight pump emits an alarm when a TBR is finished. This button lets you enable or disable this alarm without the need for configuration software.
Configure alarms in the pump as follows:
- Menu > Settings > Device settings > Mode settings > Quiet > Signal > Sound
- Menu > Settings > Device settings > Mode settings > Quiet > Volume > 0 (remove all bars)
- Menu > Modes > Signal mode > Quiet
This will silence all alarms from the pump, allowing AndroidAPS to decide if an alarm is relevant to you. If AndroidAPS does not acknowledge an alarm, its volume will increase (first beep, then vibration).
Depending on the firmware version of your pump, the Insight will vibrate briefly every time a bolus is delivered (for example, when AndroidAPS issues an SMB or TBR emulation delivers an extended bolus).
- Firmware 1.x: No vibration by design.
- Firmware 2.x: Vibration cannot be disabled.
- Firmware 3.x: AndroidAPS delivers bolus silently. (minimum version 18.104.22.168)
Firmware version can be found in the menu.
Battery life for Insight when looping range from 10 to 14 days, max. 20 days. The user reporting this is using Energizer lithium batteries.
The Insight pump has a small internal battery to keep essential functions like the clock running while you are changing the removable battery. If changing the battery takes too long, this internal battery may run out of power, the clock will reset, and you will be asked to enter a new time and date after inserting a new battery. If this happens, all entries in AndroidAPS prior to the battery change will no longer be included in calculations as the correct time cannot be identified properly.
Insight specific errors¶
Just use one extended bolus at a time as multiple extended boluses at the same time might cause errors.
Sometimes it might happen that the Insight pump does not answer during connection setup. In this case AAPS will display the following message: “Timeout during handshake - reset bluetooth”.
In this case turn off bluetooth on pump AND smartphone for about 10 seconds and then turn it back on.