세팅 설정에 따라 화면 위쪽의 탭 또는 ≡ 버튼에서 구성 관리자 메뉴에 들어갈 수 있습니다.
구성 관리자 (Conf) 탭에서 AndroidAPS의 선택 사항들을 설정 할 수 있습니다. 좌측의 박스 (A) 를 체크하면 사용하고자 하는 기능을 활성화 할 수 있습니다. 우측의 박스 (C) 를 체크하면 이 기능을 탭 (E) 메뉴로 볼 수 있습니다. 우측 박스를 체크하지 않은 경우, 이 기능을 사용하기 위해 화면 좌측 상단에 있는 ≡ 버튼 (D) 를 이용할 수 있습니다.
추가 세팅을 할 수 있는 기능들에는 톱니바퀴 버튼 (B) 이 있고, 이 버튼을 클릭하면 특정 세팅들을 할 수 있는 설정 메뉴로 연결됩니다.
초기 설정: AAPS 2.0 버전부터는 설정 마법사가 AndroidAPS 설정 과정을 도와줍니다. 이를 사용하려면 화면 우측 상단의 점 3개 메뉴 (F) 의 ‘설치 마법사’를 선택해주십시오.
Select the basal profile you wish to use. See Profiles page for more setup information.
As of AAPS 3.0, only the local profile is available.
However, it is possible to synchronise a Nightscout profile into a local profile. To do this, however, it is important to clone the whole database record consisting of several profiles in the Nightscout editor. Please see the instructions below. This can be helpful if major changes to a more extensive profile can be entered more easily via the web interface, e.g. to manually copy data from a spreadsheet.
Local profile uses the basal profile manually entered in phone. As soon as it is selected, a new tab appears in AAPS, where you can change the profile data read out from the pump if necessary. With the next profile switch they are then written to the pump in profile 1. This profile is recommended as it does not rely on internet connectivity.
Your local profiles are part of exported settings. So make sure to have a backup in a safe place.
green plus: add
red X: delete
blue arrow: duplicate
If you make any changes to your profile, make sure, you are editing the correct profile. In profile tab there is not always shown the actual profile being used - e.g. if you made a profile switch by using the profile tab on homescreen it may differ from the profile actually shown in profile tab as there is no connection between these.
프로파일 변경 사항을 복제하기¶
You can easily create a new local profile from a profile switch. In this case timeshift and percentage will be applied to the new local profile.
Click 3-dots-menu in upper right corner.
Press star symbol to access profile switch page.
Select the desired profile switch and press “Clone”.
You can edit the new local profile in Local Profile (LP) tab or via the hamburger menu.
로컬 프로파일을 Nightscout에 업로드하기¶
Local profiles can also be uploaded to Nightscout. The settings can be found in NSClient preferences.
Change profile in Nighscout profile editor¶
You can synchronoze changes to the profile in the Nighscout profile editor to local profiles. The settings can be found in NSClient preferences.
It is necessary to clone the actual active entire Nightscout database records for the profiles and not just a profile with the blue arrow! The new database records then carries the current date and can be activated via the tab “local profile”.
Profile helper offers two functions:
Find a profile for kids
Compare two profiles or profile switches in order to clone a new profile
Details are explained on the separate profile helper page.
Select the type of insulin curve you are using.
The options ‘Rapid-Acting Oref’, Ultra-Rapid Oref’, ‘Lyumjev’ and ‘Free-Peak Oref’ all have an exponential shape. More information is listed in the OpenAPS docs.
The curves will vary based on the DIA and the time to peak.
PURPLE line shows how much insulin remains after it has been injected as it decays with time.
BLUE line shows how active insulin is.
The DIA is not the same for each person. That’s why you have to test it for yourself.
But it must always be at least 5 hours.
For a lot of people using ultra-rapid insulins like Fiasp there is practically no noticeable effect after 3-4 hours any more, even if 0.0xx units are available as a rule then. This residual amount can still be noticeable during sports, for example. Therefore, AndroidAPS uses minimum 5h as DIA.
You can read more about that in the Insulin Profile section of this page.
Insulin type differences¶
For ‘Rapid-Acting’, ‘Ultra-Rapid’ and ‘Lyumjev’ the DIA is the only variable you can adjust by yourself, the time to peak is fixed.
Free-Peak allows you to adjust both the DIA and the time to peak, and must only be used by advanced users who know the effects of these settings.
The insulin curve graph helps you to understand the different curves.
You can view it by enabling the tickbox to show it as a tab, otherwise it will be in the hamburger menu.
recommended for Humalog, Novolog and Novorapid
DIA = at least 5.0h
Max. peak = 75 minutes after injection (fixed, not adjustable)
recommended for FIASP
DIA = at least 5.0h
Max. peak = 55 minutes after injection (fixed, not adjustable)
special insulin profile for Lyumjev
DIA = at least 5.0h
Max. peak = 45 minutes after injection (fixed, not adjustable)
Free Peak Oref¶
With the “Free Peak 0ref” profile you can individually enter the peak time. To do so click to cogwheel to enter advanced settings.
The DIA is automatically set to 5 hours if it is not specified higher in the profile.
This effect profile is recommended if an unbacked insulin or a mixture of different insulins is used.
Select the blood glucose source you are using - see BG Source page for more setup information.
xDrip+ - Cannot be used as receiver for Dexcom G6 as of AAPS 3.0 (see release notes for details.
Glimp - only version 4.15.57 and newer are supported
Tomato App for MiaoMiao device
Glunovo App for Glunovo CGM system
NSClient BG - not recommended as closed loop relies on mobile data / wifi coverage in this case. CGM data will only be received if there is an online connection to your NS site. Better use local broadcast from one of the other CGM data sources.
Random BG: Generates random BG data (Demo mode only)
Select the pump you are using.
Dana R Korean (for domestic DanaR pump)
Dana Rv2 (DanaR pump with unofficial firmware upgrade)
For dana pumps, use Advanced settings to activate BT watchdog if necessary. It switches off bluetooth for one second if no connection to the pump is possible. This may help on some phones where the bluetooth stack freezes.
Password for Dana RS pump must be entered correctly. Password was not checked in previous versions.
Accu Chek Combo (requires ruffy installation)
MDI (receive AAPS suggestions for your multiple daily injections therapy)
Virtual pump (open loop for pump which don’t have any driver yet - AAPS suggestions only)
Select the type of sensitivity detection. For more details of different designs please read on here. This will analyze historical data on the go and make adjustments if it recognizes that you are reacting more sensitively (or conversely, more resistant) to insulin than usual. More details about the Sensitivity algorithm can be read in the OpenAPS docs.
You can view your sensitivity on the homescreen by selecting SEN and watching the white line. Note, you need to be in Objective 8 in order to let Sensitivity Detection/Autosens automatically adjust the amount of insulin delivered. Before reaching that objective, the Autosens percentage / the line in your graph is displayed for information only.
If you use Oref1 with SMB you must change min_5m_carbimpact to 8. The value is only used during gaps in CGM readings or when physical activity “uses up” all the blood glucose rise that would otherwise cause AAPS to decay COB. At times when carb absorption can’t be dynamically worked out based on your bloods reactions it inserts a default decay to your carbs. Basically, it is a failsafe.
Select the desired APS algorithm for therapy adjustments. You can view the active detail of the chosen algorithm in the OpenAPS(OAPS) tab.
OpenAPS AMA (advanced meal assist, state of the algorithm in 2017) In simple terms the benefits are after you give yourself a meal bolus the system can high-temp more quickly IF you enter carbs reliably.
OpenAPS SMB (super micro bolus, most recent algorithm for advanced users) Note you need to be in Objective 9 in order to use OpenAPS SMB and min_5m_carbimpact must be set to 8 in Config builder > Sensitivity detection > Sensitivity Oref1 settings.
Switch between Open Loop, Closed Loop and Low Glucose Suspend (LGS).
AAPS continuously evaluates all available data (IOB, COB, BG…) and makes treatment suggestions on how to adjust your therapy if necessary.
The suggestions will not be executed automatically (as in closed loop) have to be entered manually into the pump or by using a button in case you are using a compatible pump (Dana R/RS or Accu Chek Combo).
This option is for getting to know how AndroidAPS works or if you are using an unsupported pump.
AAPS continuously evaluates all available data (IOB, COB, BG…) and automatically adjusts the treatment if necessary (i.e. without further intervention by you) to reach the set target range or value (bolus delivery, temporary basal rate, insulin switch-off to avoid hypo etc.).
The Closed Loop works within numerous safety limits, which you can be set individually.
Closed Loop is only possible if you are in Objective 6 or higher and use a supported pump.
Please note: In closed loop mode a single target instead of target range (i.e. 5,5 mmol or 100 mg/dl instead of 5,0 - 7,0 mmol or 90 - 125 mg/dl) is recommended.
Low Glucose Suspend (LGS)¶
maxIOB is set to zero
This means if blood glucose is dropping it can reduce basal for you.
But if blood glucose is rising no automatic correction will be made. Your basal rates will remain the same as your selected profile.
Only if basal IOB is negative (from a previous Low Glucose Suspend) additional insulin will be given to lower BG.
Minimal request change¶
When using open loop you will receive notifications every time AAPS recommends to adjust basal rate.
To reduce number of notifications you can either use a wider bg target range or increase percentage of the minimal request rate.
This defines the relative change required to trigger a notification.
목표 (학습 프로그램)¶
AndroidAPS has a leraning program (objectives) that you have to fulfill step by step. This should guide you safely through setting up a closed loop system. It guarantees that you have set everything up correctly and understand what the system does exactly. This is the only way you can trust the system.
You should export your settings (including progress of the objectives) on a regularly basis. In case you have to replace your smartphone later (new purchase, display damage etc.) you can simply import those settings.
See Objectives page for more information.
If you view the Treatments (Treat) tab, you can see the treatments that have been uploaded to nightscout. Should you wish to edit or delete an entry (e.g. you ate less carbs than you expected) then select ‘Remove’ and enter the new value (change the time if necessary) through the carbs button on the home screen.
Displays the current state of your loop and buttons for most common actions (see section The Homescreen for details). Settings can be accessed by clicking the cog wheel.
화면을 켜진 상태로 유지¶
Option ‘Keep screen on’ will force Android to keep the screen on at all times. This is useful for presentations etc. But it consumes a lot of battery power. Therefore, it is recommended to connect the smartphone to a charger cable.
Define which Buttons are shown on the home screen.
CGM (opens xDrip+)
Furthermore, you can set shortcuts for insulin and carb increments and decide whether the notes field should be shown in treatment dialogues.
Create a button for a certain standard meal (carbs and calculation method for the bolus) which will be displayed on the home screen. Use for standard meals frequently eaten. If different times are specified for the different meals you will always have the appropriate standard meal button on the home screen, depending on the time of day.
Note: Button will not be visible if outside the specified time range or if you have enough IOB to cover the carbs defined in the QuickWizard button.
Choose default temp-targets (duration and target). Preset values are:
eating soon: target 72 mg/dl / 4.0 mmol/l, duration 45 min
activity: target 140 mg/dl / 7.8 mmol/l, duration 90 min
hypo: target 125 mg/dl / 6.9 mmol/l, duration 45 min
Fill/Prime standard insulin amounts¶
Choose the default amounts of the three buttons in fill/prime dialogue, depending on the length of your catheter.
Range of visualization¶
Choose the high and low marks for the BG-graph on AndroidAPS overview and smart watch. It is only the visualization, not the target range for your BG. Example: 70 - 180 mg/dl or 3.9 - 10 mmol/l
Shorten tab titles¶
Choose wether the tab titles in AndroidAPS are long (e.g. ACTIONS, LOCAL PROFILE, AUTOMATION) or short (e.g. ACT, LP, AUTO)
Show notes field in treatment dialogs¶
Choose if you want to have a notes field when entering treatments or not.
Choose if you want to have status lights on overview for cannula age, insulin age, sensor age, battery age, reservoir level or battery level. When warning level is reached, the color of the status light will switch to yellow. Critical age will show up in red.
Deliver this part of bolus wizard result: When using SMB, many people do not meal-bolus 100% of needed insulin, but only a part of it (e.g. 75 %) and let the SMB with UAM (unattended meal detection) do the rest. In this setting, you can choose a default value for the percenteage the bolus wizard should calculate with. If this setting is 75 % and you had to bolus 10u, the bolus wizard will propose a meal bolus of only 7.5 units.
Enable super bolus functionality in wizard (It is different from super micro bolus!): Use with caution and do not enable until you learn what it really does. Basically, the basal for the next two hours is added to the bolus and a two hour zero-temp activated. AAPS looping functions will be disabled - so use with care! If you use SMB AAPS looping functions will be disabled according to your settings in “Max minutes of basal to limit SMB to”, if you do not use SMB looping functions will be disabled for two hours. Details on super bolus can be found here.
Some buttons to quickly access common features.
See AAPS screenshots for details.
User defined automation tasks (‘if-then-else’). Please read on here.
Allows remote caregivers to control some AndroidAPS features via SMS, see SMS Commands for more setup information.
Displays the food presets defined in the Nightscout food database, see Nightscout Readme for more setup information.
Note: Entries cannot be used in the AndroidAPS calculator. (View only)
Monitor and control AAPS using your Android Wear watch (see page Watchfaces). Use settings (cog wheel) to define which variables should be considered when calculating bolus given though your watch (i.e. 15min trend, COB…).
Watch에서 bolus 주입 등을 하려면, “Wear 설정”에서 “Watch로부터 컨트롤하기”를 사용합니다.
Through Wear tab or hamburger menu (top left of screen, if tab is not displayed) you can
Resend all data. Might be helpful if watch was not connected for some time and you want to push the information to the watch.
Open settings on your watch directly from your phone.
xDrip Statusline (watch)¶
Display loop information on your xDrip+ watchface (if you are not using AAPS/AAPSv2 watchface
Setup sync of your AndroidAPS data with Nightscout.
Settings in preferences can be opened by clicking the cog wheel.
Email and number of logs to be send. Normally no change necessary.
Use tab for config builder instead of hamburger menu.