Documentation about profiles in general can be found at Config Builder - profile.
On starting your AAPS and selecting your profile, you will need to do a “Profile switch” event with zero duration (explained later). By doing this AAPS starts tracking history of profiles and every new profile change requires another “Profile switch” even when you change content of the profile in NS. Updated profile is pushed to AAPS immediately, but you need to switch the same profile again to start using these changes.
Internally AAPS creates snapshot of profile with start date and duration and is using it within selected period.
- Duration of zero means infinite. Such profile is valid until new “Profile switch”.
- Duration of x minutes means x minutes use of this profile. After that duration the profile is switched back to the previous valid “Profile switch”.
If you edited your profile inside the “local profile” tab you can activate the profile there which makes an implicit profile switch too.
To do a profile switch long-press on the name of your profile (“Tuned 03/11” in the picture below) on the homescreen of AndroidAPS.
Keisdami profilius, galite pasirinkti dvi papildomas parinktis, kurios anksčiau buvo Cirkadinio procento profilio dalis:
This applies the same percentage to all parameters.
If you set it to 130% (meaning you are 30% more insulin resistant), it will raise the basal rate by 30%. It will also lower the ISF and IC accordingly (divide by 1.3 in this example).
It will be sent to the pump and then be the default basal rate.
The loop algorithm (open or closed) will continue to work on top of the selected percentage profile. So, for example separate percentage profiles can be set up for different stages of the hormone cycle.
- This moves everything round the clock by the number of hours entered.
- So, for example, when working night shifts change the number of hours to how much later/earlier you go to bed or wake up.
- It is always a question of which hour’s profile settings should replace the settings of the current time. This time must be shifted by x hours. So be aware of the directions as described in the following example:
- Current time: 12:00
- Positive time shift
- 2:00 +10 h -> 12:00
- Settings from 2:00 will be used instead of the settings normally used at 12:00 because of the positive time shift.
- Negative time shift
- 22:00 -10 h -> 12:00
- Settings from 22:00 (10 pm) will be used instead of the settings normally used at 12:00 because of the negative time shift.
Šis profilio fiksavimo mechanizmas leidžia daug tikslesnius praeities skaičiavimus ir taip pat leidžia stebėti profilio pokyčius.
Profilio klaidų šalinimas¶
“Neteisingas profilis” / “Pagrindinis profilis nesutampa su valandomis”¶
These error messages will appear if you have any basal rates or I:C rates not on the hour. (DanaR and DanaRS pumps do not support changes on the half hour for example.)
Remember / note down date and time shown in the error message (26/07/2019 5:45 pm in screenshot above).
Go to Treatments tab
Scroll until you find date and time from error message.
Use remove function.
Sometimes there is not only one faulty profile switch. In this case remove also the others.
Kaip alternatyvą, galite ištrinti profilio pakeitimą tiesiogiai mLab, kaip aprašyta toliau.
„Profilio pakeitimas gautas iš NS, tačiau profilis neegzistuoja lokaliai“¶
- The requested profile was not synced correctly from Nightscout.
- Follow instructions from above to delete the profile switch
Kaip alternatyvą, galite ištrinti profilio keitimą tiesiogiai mLab:
- Go to your mlab collection
- Search in the treatments for profile switch
- Delete the profile switch with date and time that was mentioned in the error message.