Filteren van bloed glucose waardes
If BG data is jumpy/noisy, AAPS may dose insulin incorrectly resulting in high or low BG. For this reason it’s important to disable the loop until the problem is resolved. Depending on your CGM such issues may be due to the CGM’s configuration or sensor problems/site issues. You may need to replace your CGM sensor to resolve this.
Some CGM systems have internal algorithms to detect the noise level in the readings and AndroidAPS can use this information to avoid giving SMBs if the BG data is too unreliable. However, some CGMs do not transmit this data and for these BG sources ‘Enable SMB always’ and ‘Enable SMB after carbs’ are disabled for safety reasons.
Build Your Own Dexcom App
When using BYODA your BG data is smooth and consistent. Furthermore you can take advantage of Dexcom back-smoothing. There are no restrictions in using SMBs, because the noise-level data is shared with AAPS.
xDrip+ with Dexcom G6 or Dexcom ONE
Noise-level data and smooth BG readings are only shared with AAPS if you use xDrip+ native mode. Using native mode, there are no restrictions in using SMBs.
Dexcom G6 or Dexcom ONE with xDrip+ Companion Mode
The noise-level data is not shared with AAPS using this method. Therefore ‘Enable SMB always’ and ‘Enable SMB after carbs’ are disabled.
Freestyle Libre sensors
xDrip+ with FreeStyle Libre
None of the FreeStyle Libre systems (FSL1, FSL2, or FSL3) broadcast any information about the level of noise detected in the readings, and therefore ‘Enable SMB always’ and ‘Enable SMB after carbs’ are disabled for all setups using the FreeStyle Libre.
In addition, many people have reported the FreeStyle Libre often produces noisy data. In xDrip+ there are a few options to help with this:
Smooth Sensor Noise. In xDrip+ Settings > xDrip+ Display Settings ensure that Smooth Sensor Noise is turned on. This attempts to apply smoothing to noisy data.
Smooth Sensor Noise (Ultrasensitive). If you are still seeing noisy data in xDrip+ you can apply more aggressive smoothing using the Smooth Sensor Noise (Ultrasensitive) setting. This will attempt to apply smoothing even on very low levels of detected noise. To do this, first enable engineering mode in xDrip+. Then navigate to Settings > xDrip+ Display Settings and turn on Smooth Sensor Noise (Ultrasensitive).